# Multimeter / Digital Multimeter

A multimeter or digital multimeter (DMM) is a gadget that gives a few sorts of estimations inside an instrument. DMM utilizes a more precise and lucid computerized yield than a simple arrangement. Without utilizing DMM to rapidly quantify live voltages and flows, there is a danger of horrendous stun that can likewise be deadly. Accordingly, it represents a genuine security danger.

We offer a genuine RMS digital multimeter that has manual or programmed go winnow and with or without an RS-232 interface for moving information to a PC. The comparing test links for the digital multimeter accompany the invention just as batteries with the goal that it is yare to use straight out the container. The digital multimeter can be joined by an ISO alignment testament either when it is injunctively approved or included as a part of a yearly recalibration.

Present-day multimeters are frequently digital because of their exactness, solidness, and extra highlights. In an advanced multimeter, the sign under test is changed over into a voltage and sign to a speaker with electronically-controlled addition. A digital multimeter shows the amount estimated as a number, which dispenses with parallax mistakes.

Digital Multimeter Suppliers in Jharkhand | Digital Multimeter Suppliers in Bihar | Digital Multimeter Suppliers in India | Multimeter | Digital Multimeter Price | Multimeter Price

Atomic Absorption Spectrophotometer | Digital Multimeter | Concrete Moisture Meter | Gloss Meter | Gas Detector | High-Speed Camera | Infrared Thermometer | Microscope | pH Meter | Particle Counter | Power Supply Unit | Paperless Recorder | Spectrophotometer | Torque Meter | Thermometer | Viscometer | Water Analysis Meter

Another special feature of the digital multimeter is the PASS / FAIL function. With this function, target ranges for the measured values can be set for each measuring function. Depending on the actual measured value, a "PASS" (passed) or a "FAIL" (failed) is then shown on the display by the digital multimeter. The digital multimeter is therefore used, for example, in quality control. With the help of the adjustable handle, the digital multimeter can be adjusted for ergonomic use by the operator. The handle of the digital multimeter can also be used to change the workplace. The housing of the digital multimeter has a rubberized edge protection, which prevents damage.

- Table multimeter with reinforced housing

- USB interface for PC connection

- PASS / FAIL function for quality control

- extensive measuring functions

- Relate the measured value

- Measured value with bar graph display

Specifications

AC voltage

Measuring range Resolution Accuracy

60 mV 0.001 mV ± (0.6% 60 digits) @ 45 Hz ... 1 kHz

600 mV 0.01 mV ± (0.3% 30 digits) @ 45 Hz ... 1 kHz

6V 0.0001V ± (0.3% 30 digits) @ 45 Hz ... 1 kHz

60V 0.001V ± (0.3% 30 digits) @ 45 Hz ... 1 kHz

600V 0.01V ± (0.4% 30 digits) @ 45 Hz ... 1 kHz

1000V 0.1V ± (0.6% 30 digits) @ 45 Hz ... 1 kHz

Input impedance:> 10MΩ

Surge protection: 1000V

Display: True RMS in the measuring range of 10 ... 100% of the respective measuring range

DC voltage

Measuring range Resolution Accuracy

60 mV 0.001 mV ± (0.025% + 20 digits)

600 mV 0.01 mV ± (0.025% + 5 digits)

6V 0.0001V ± (0.025% + 5 digits)

60V 0.001V ± (0.025% + 5 digits)

600V 0.01V ± (0.003% + 5 digits)

1000V 0.1V ± (0.003% + 5 digits)

Input impedance:> 10MΩ

Surge protection: 1000V

AC voltage + DC voltage

Measuring range Resolution Accuracy

60 mV 0.001 mV ± (1% + 80 digits) @ 50 Hz ... 1 kHz

600 mV 0.01 mV ± (1% + 80 digits) @ 50 Hz ... 1 kHz

6V 0.0001V ± (1% + 80 digits) @ 50 Hz ... 1 kHz

60V 0.001V ± (1% + 80 digits) @ 50 Hz ... 1 kHz

600V 0.01V ± (1% + 80 digits) @ 50 Hz ... 1 kHz

1000V 0.1V ± (1.2% + 80 digits) @ 50 Hz ... 1 kHz

Input impedance:> 10MΩ

Surge protection: 1000V

Display: True RMS in the measuring range of 10 ... 100% of the respective measuring range

Alternating current

Measuring range Resolution Accuracy

600 µA 0.01 µA ± (0.6% + 40 digits) @ 45 Hz ... 1 kHz

6000 µA 0.1 µA ± (0.6% + 20 digits) @ 45 Hz ... 1 kHz

60-mA 0.001-mA ± (0.6% + 40 digits) @ 45 Hz ... 1 kHz

600-mA 0.01-mA ± (0.6% + 20 digits) @ 45 Hz ... 1 kHz

10 A 0.001 A ± (1% + 20 digits) @ 45 Hz ... 1 kHz

Display: True RMS in the measuring range of 10 ... 100% of the respective measuring range

Overload protection: µA and mA Measurement: Fuse F 0.6 A 1000V Ø6 x 32 mm

10 A measurement: F 11 A 1000V Ø 10 x 38 mm

When measuring almost 20 A, the measurement time should not be longer than 30 seconds.

After the measurement, the measuring device must rest for approx. 10 minutes.

Direct current

Measuring range Resolution Accuracy

600 µA 0.01 µA ± (0.8% + 20 digits)

6000 µA 0.1 µA ± (0.8% + 10 digits)

60-mA 0.001-mA ± (0.8% + 20 digits)

600-mA 0.01-mA ± (0.15% + 10 digits)

10 A 0.001 A ± (0.5% + 10 digits)

Overload protection: µA and mA Measurement: Fuse F 0.6 A 1000V Ø6 x 32 mm

10 A measurement: F 11 A 1000V Ø10 x 38 mm

A measurement of almost 20 A should not take longer than 30 seconds.

After the measurement, the measuring device must rest for approx. 10 minutes.

AC + DC

Measuring range Resolution Accuracy

600 µA 0.01 µA ± (0.8% + 40 digits) @ 50 Hz ... 1 kHz

6000 µA 0.1 µA ± (0.8% + 20 digits) @ 50 Hz ... 1 kHz

60-mA 0.001-mA ± (0.8% + 40 digits) @ 50 Hz ... 1 kHz

600-mA 0.01-mA ± (0.8% + 20 digits) @ 50 Hz ... 1 kHz

10 A 0.001 A ± (1.2% + 20 digits) @ 50 Hz ... 1 kHz

Display: True RMS in the measuring range of 10 ... 100% of the respective measuring range

Overload protection: µA and mA Measurement: Fuse F 0.6 A 1000V Ø6 x 32 mm

10 A measurement: F 11 A 1000V Ø10 x 38 mm

A measurement of almost 20 A should not take longer than 30 seconds.

After the measurement, the measuring device must rest for approx. 10 minutes.

Resistance

Measuring range Resolution Accuracy

600 Ω 0.01Ω In REL mode: ± (0.05% + 10 digits)

6 kΩ 0.0001 kΩ ± (0.05% + 2 digits)

60 kΩ 0.001 kΩ ± (0.05% + 2 digits)

600 kΩ 0.01 kΩ ± (0.05% + 2 digits)

6 MΩ 0.0001 MΩ ± (0.3% + 10 digits)

60 MΩ 0.001 MΩ ± (2% +10 digits)

Surge protection: 1000 V

With a measuring range of 60 MΩ the ambient humidity should be < 50%.

Conductivity

Measuring range Resolution Accuracy

60 nS 0.01 nS ± (2% + 10 digits)

Surge protection: 1000V

The ambient humidity for the measuring range should be < 50%.

Capacity measurement

Measuring range Resolution Accuracy

6 nF 0.001 nF ± (3% + 10 digits)

60 nF 0.01 nF ± (2.5% + 5 digits)

600 nF 0.1 nF ± (2% + 5 digits)

6 μF 0.001 μF ± (2% + 5 digits)

60 μF 0.01 μF ± (2% + 5 digits)

600 μF 0.1 μF ± (2% + 5 digits)

6 mF 1 μF ± (5% + 5 digits)

60 mF 10 μF for reference only

Surge protection: 1000 V

Temperature

Measuring range Resolution Accuracy

-40 ... 40°C 1°C ± (2.0% + 30 digits)

40 ... 400°C 1°C ± (1.0% + 30 digits)

100 ... 1000°C 1°C ± 2.5%

-40 ... 104°F 1.8°F ± (2.5% + 50 digits)

104 ... 752°F 1.8°F ± (1.5% + 50 digits)

752 ... 1832°F 1.8°F ± 2.5%

Surge protection: 1000V

Two-channel temperature measurement

Temperature sensor: Type K only applies to the measurement of temperatures below 230°C / 446°F

Frequency measurement

Measuring range Resolution Accuracy

60 Hz 0.001 Hz ± (0.02% + 8 digits)

600 Hz 0.01 Hz ± (0.01% + 5 digits)

6 kHz 0.0001 kHz ± (0.01% + 5 digits)

60 kHz 0.00 1kHz ± (0.01% + 5 digits)

600 kHz 0.01 kHz ± (0.01% + 5 digits)

6 MHz 0.0001 MHz ± (0.01% + 5 digits)

60 MHz 0.00 1MHz ± (0.01% + 5 digits)

Surge protection: 1000V

Input amplitude: 10 Hz ... 30 MHz: 600 mV <a <30 Vrm,> 30 MHz: not specified

Duty cycle

Measuring range Resolution Accuracy

10 ... 90% @ 0.01% ± (1.2% + 30 digits)

(10Hz ... 2kHz)

Surge protection: 1000V

Rise time: <1 µs, the signal is centered on the trigger level

Pulse width

Measuring range Resolution Accuracy

250 mS 0.001 ... 0.1 mS ± (1.2% + 30 digits)

Surge protection: 1000V

Rise time: < 1 µs, the signal is centered on the trigger level

10 Hz ... 200 kHz: pulse width > 2 µs

The pulse range is limited by the frequency of the signal

Continuity test

Resolution Functional description

0.01 Ω Short circuit alarm: from < 10 Ω, alarm switches off from > 50 Ω

Surge protection: 1000V

Diode test

0.0001V The open circuit voltage is approx. 3V and the measurable voltage drop

of the PN transition is < 3V A continuous beep

indicates the short circuit of the PN junction. The typical tension

of the silicon PN junction is between 0.5 ... 0.8V.

Surge protection: 1000V

Further specifications on accuracy can be found in the operating instructions

Further specification

Interface USB interface

Supply voltage 100 ... 240V adjustable

Environmental conditions 23°C / 73.4°F ± 5°C / 41°F, < 75% RH

Dimensions 310 x 240 x 120 mm / 12.2 x 9.4 x 4.7 in

Weight Approx. 3713 g / 8.2 lbs

Delivery Scope

1 x Digital multimeter PCE-BDM 20

1 x USB cable

1 x Connection adapter for thermocouples

2 x Thermocouple type K x software

2 x Crocodile clip

1 x IEC cable

1 x Instruction manual

In addition, the measured values can be stored on an SD card of up to 32 GB on the table multimeter using the data logger function. As a result, long-term recordings of electrical components or assemblies, machinery and equipment are possible. Thanks to the large and illuminated display, the readings are very easy to read. This table multimeter can optionally also be laboratory-calibrated and equipped with an ISO calibration certificate (at the time of initial order or recalibration, eg annually).

- Automatic / manual range selection

- Also suitable for mobile use

- Continuity test, diode test

- TrueRMS measurement

- RS-232 interface

- Mains / AC power adapter and battery operation

- Frequency measurement up to 60 MHz

- ISO calibration optionally available

Specification

Measuring range DC

Measuring ranges 600.0mV / 6V / 60V / 600V / 1000V

Resolution 0.1mV / 0.001V / 0.01V / 0.1V / 1V

Accuracy ± (0.5% + 2 d) to 600 mV

± (0.8% + 1 d) from 600 mV

Input resistance 10 Mohms

Overvoltage protection In the measuring range 600 mV to ± 350 VAC / VDC

over measuring range 600 mV to ± 1000 VAC / VDC

Measuring range AC

Measuring ranges 600.0mV / 6V / 60V / 600V / 1000V

Resolution 0.1mV / 0.001V / 0.01V / 0.1V / 1V

Accuracy ± (1% + 3 d) at a frequency of 50/60 Hz

Input resistance 10 MΩ

Overvoltage protection in the measuring range 600 mV to ± 350 VAC / VDC

over measuring range 600 mV to ± 1000 VAC / VDC

direct current

measuring range Resolution Accuracy Fuse

10 A 0.01A ± (1.5% + 2 Dgt) 10 A / 600V

6 A 0.001 A ± (1.5% + 5 Dgt) 10 A / 600V

600-mA 0.1-mA ± (0.5% + 2 Dgt) 600 mA / 600V

60-mA 0.01-mA ± (0.5% + 2 Dgt) 600 mA / 600V

6000 μA 1 μA ± (0.5% + 2 Dgt) 600 mA / 600V

600 μA 0.1 μA ± (0.5% + 2 Dgt) 600 mA / 600V

Alternating current

Measuring range Resolution Accuracy Fuse

10 A 0.01A ± (1.5% + 2 Dgt) 10 A / 600V

6 A 0.001 A ± (1.5% + 5 Dgt 10 A / 600V

60-mA 0.1-mA ± (1% + 7 Dgt) 600-mA / 600V

600-mA 0.01-mA ± (1% + 7 Dgt) 600-mA / 600V

6000 μA 1 μA ± (1% + 7 Dgt 600-mA / 600V

600 μA 0.1 μA ± (1% + 7 Dgt) 600-mA / 600V

The accuracies refer to 50 and 60 Hz

Diode test

Measuring range 2.7 VDC

Accuracy ± (0.5% + 2 Dgt)

Frequency measurement

Area Resolution Accuracy

60 MHz 0.01 MHz ± (0.5% + 2 Dgt)

6 MHz 0.001 MHz ± (0.5% + 2 Dgt)

600 KHz 0.1 KHz ± (0.5% + 2 Dgt)

60 KHz 0.01 KHz ± (0.5% + 2 Dgt

6 KHz 0.001 KHz ± (0.5% + 2 Dgt)

600 Hz 0.1 Hz ± (0.5% + 2 Dgt)

60 Hz 0.01 Hz ± (0.5% + 2 Dgt)

Sensitivity Min. 1 V rms, Max. 5 V rms

Continuity measurement

Acoustic signal with a resistance of less than 3 Ω

Duty cycle

Frequency range Duty cycle range

60 Hz ... 600 Hz 5 ... 90%

601 Hz ... 6 kHz 10 ... 90%

6.1 kHz ... 60 KHz 20 ... 80%

61 KHz ... 1 MHz 30 ... 80%

> 1 MHz ... 10 MH Only as a reference measurement

Accuracy ± (0.5% + 5 Dgt)

Sensitivity Min. 1V rms, Max. 5V rms

Resistivity

Measuring range Resolution Accuracy Fuse

60 MΩ 0.01 MΩ ± (3% + 5 Dgt) ± 350 VAC / DC

6 MΩ 0.001 MΩ ± (1.5% + 2 Dgt) ± 350 VAC / DC

600 kΩ 0.1 kΩ ± (1.5% + 2 Dgt) ± 350 VAC / DC

60 kΩ 0,01kΩ ± (1.5% + 2 Dgt) ± 350 VAC / DC

6 kΩ 0.001 kΩ ± (1.5% + 2 Dgt ± 350 VAC / DC

600 Ω 0.1 Ω ± (1% + 2 Dgt) ± 350 VAC / DC

Display 97 mm x 56 mm / 3.8 x 2.2 in

Backlit LC display up to 6000

Display updating Average between 0.5 ... 1 second

Storage rate 0, 1, 2, 5, 10, 30, 60, 120, 300, 600, 1800, 3600

*a storage rate of 0 means manual storage

Corrupt data <0.1% of the data is typically faulty

SD card capacity 4 ... 32 GB

Select measuring range Automatic and manual

Additional function Put the measured value in relation

Freeze measured value

Automatic shutdown

MAX / MIN

Polarity In reverse polarity, the measured value is negated.

Zero Automatically

Interface RS232

Power supply power supply Primary: 230V, 50 Hz, 0.3 A

Secondary: 9 VDC, 800-mA, 7.2 VA

Power supply Batteries 6 x AA 1.5V

Dimensions 292 x 236 x 98 mm /

11.5 x 9.3 x 3.9 in

Weight 1972 g / 4.3 lbs (without batteries)

Environmental conditions 0 ... 50°C / 32 ... 122°F, max. 80% RH

Degree of protection / Standardization CAT I 1000V

Delivery Scope

1 x Digital Multimeter PCE-BMM 10

1 x Set of test leads

1 x SD card

1 x User manual

With the LAN tester, which also serves as a multimeter, it is possible to perform a continuity test with buzzer and a diode test. The LAN tester is also a device that identifies the most common network installation errors. The two-row LEDs successively indicate the pin assignment of the cable to be tested. In the event of a short circuit, more than two LEDs light up simultaneously.

- Combination device of LAN tester and multimeter

- LCD

- Checks Lankabel and patch fields

- Continuity test and diode test possible

- Overload indicator

- Short circuit display

Specifications

Measuring function AC Resolution

Measuring range 400 mV 0.1 mV

4V 1 mV

40V 10 mV

400V 100 mV

750V 1V

Accuracy ± (1% from + 5 Dgt)

Measuring function DC Resolution

Measuring range 400 mV 0.1 mV

4 V 1 mV

40V 10 mV

400V 100 mV

1000V 1V

Accuracy ± (0.5% of v / m + 3Dgt)

Measuring function Resistance Resolution

Measuring range 400 Ω 0.1 Ω

4 kΩ 1 Ω

40 kΩ 10 Ω

400 kΩ 100 Ω

4 MΩ 1 kΩ

Accuracy ± (1.2% from + 3 Dgt)

Resistance Resolution

40 MΩ 10 kΩ

Accuracy ± (2% from + 4 Dgt)

Continuity test 400 Ω

Alarm Measured value <25Ω

Display LED bargraph for LAN testers

LC display for multimeter functions

Power supply LAN tester: 1 x 9V 6F22 battery

Multimeter: 2 x 1.5V AAA battery

Dimensions Handset: 165 x 82 x 42 mm / 6.5 x 3.2 x 1.7 in

Remote unit: 75 x 30 x 25 mm / 3 x 1.2 x 1 in

Weight Handset: 240 g / < 1 lb

Remote unit: 30 g / < 1 lb

Electric security CAT III 600V

Delivery Scope

1 x Multimeter PCE-LT 15

1 x Remote unit

1 x RJ45 connection cable

1 x Test lead set

2 x 1.5V AAA batteries

1 x 9V block battery

1 x User manual

The voltmeter has an NCV measuring function to quickly check whether the mains voltage is present on a line or on an electrical device. The special feature of this function with the voltmeter is that no lines have to be disconnected for the measurement. Only the voltmeter has to be held against the test object or the line.

In addition to these measurement functions, the display of the voltmeter is backlit. As a result, the measured values can be read from the voltmeter even under difficult conditions. If the measuring point is difficult to identify due to insufficient lighting due to a power failure, for example, the voltmeter with the built-in flashlight can remedy this. The rubber cover of the voltmeter also protects the measuring device from damage.

- Voltmeter, ammeter, NCV

- Small and compact design

- Backlight for a better view

- Flashlight for illumination

- Automatic and manual measuring range

- "Hold" function for freezing the measured value

Specification

DC

Measuring range Resolution Accuracy

600 mV

0.1 mV ± (0.5% + 4 digits)

6V 1 mV ± (0.5% + 4 digits)

60V 10 mV ± (0.5% + 4 digits)

600V 100 mV ± (1.0% + 4 digits)

Input impedance: > 40 MΩ for the measuring range 600 mV; 10 MΩ for all

other measuring ranges

Surge protection: 600V DC / AC rms

AC

Measuring range Resolution Accuracy

6V 1 mV ± (0.8% + 6 digits)

60V 10 mV ± (0.8% + 6 digits)

600V 100 mV ± (1.0% + 6 digits)

Input impedance: 10 MΩ

Surge protection: 600V DC / AC rms

Frequency range for the measuring range 600V: 40 ... 1000 Hz

Frequency range for all other measuring ranges: 40 ... 2000 Hz

Display: RMS of the sine wave

Direct current

Measuring range Resolution Accuracy

600 µA 0.1 µA ± (1.0% + 5 digits)

6000 µA 1 µA ± (1.0% + 5 digits)

60-mA 10 µA ± (1.0% + 5 digits)

600-mA 100 µA ± (1.0% + 5 digits)

6 A

1-mA ± (1.0% + 5 digits)

10 A 10-mA ± (2.0% + 10 digits)

Maximum voltage drop: 200 mV

Maximum current measurement: 10 A (within a time of 10 seconds)

Fuse: 0.5 A / 250V, 10 A / 250V

Alternating current

Measuring range Resolution Accuracy

600 µA 0.1 µA ± (1.5% + 5 digits)

6000 µA 1 µA ± (1.5% + 5 digits)

60-mA 10 µA ± (1.5% + 5 digits)

600-mA 100 µA ± (1.5% + 5 digits)

6 A 1-mA ± (1.5% + 5 digits)

10 A 10-mA

± (2.0% + 10 digits)

Maximum voltage drop: 200 mV

Maximum current measurement: 10 A (within a time of 10 seconds)

Fuse: 0.5 A / 250V, 10 A / 250V

Frequency range for the 10 A measuring range: 40 ... 100 Hz

Frequency range for all other measuring ranges: 40 ... 400 Hz

Capacity

Measuring range Resolution Accuracy

60 nF 10 pF ± (2.5% + 6 digits)

600 nF 100 Pf ± (2.5% + 5 digits)

6 µF 1 nF ± (2.5% + 5 digits)

60 µF 10 nF

± (2.5% + 5 digits)

600 µF 100 nF ± (5.0% + 8 digits)

6 mF 1 µF ± (5.0% + 8 digits)

60 mF 10 µF

± (5.0% + 8 digits)

Overvoltage protection: 250V DC / AC peak value

Frequency

Measuring range Resolution Accuracy

10 Hz 0.01 Hz ± (0.5% + 4 digits)

100 Hz 0.1 Hz ± (0.5% + 4 digits)

1000 Hz 1 Hz ± (0.5% + 4 digits)

10 kHz 10 Hz ± (0.5% + 4 digits)

100 kHz 100 Hz ± (0.5% + 4 digits)

1 MHz 1 kHz ± (0.5% + 4 digits)

10 MHz 10 kHz ± (0.5% + 4 digits)

Input sensitivity: 0.7V rms

Overvoltage protection: 250V DC / AC peak value

Temperature thermocouple: Type-K

Measuring range Resolution Accuracy

-40 ... 1000°C 1°C < 400°C ± (1.0% + 5 digits)

> 400°C ± (1.5% + 15 digits)

0 ... 1832°F 1°F < 752°F ± (1.0% + 5 digits)

> 752°F ± (1.5% + 15 digits)

Diode test Test current Test voltage Accuracy

About 0.8-mA Approx. 2.2V 5%

Surge protection: 600V DC / AC rms

Continuity test Test current Test voltage Accuracy

Approx. 0.4-mA Approx.3.3V 5%

Surge protection: 600V DC / AC rms

Beep at < 50 Ω

Further specification

Display LCD display

Largest display 6000 (3 5/6)

Measuring rate 3 readings per second

Display at "OIL"

Overrange

Operating conditions 0 ... 50°C / 32 ... 122°F, < 80% RH non-condensing

Storage conditions -10 ... 50°C / 14 ... 122°F, < 80% RH non-condensing

Battery indicator Display when batteries are empty

Power supply 2 x 1.5V batteries

Dimensions 140 x 72 x 37 mm / 5.5 x 2.8 x 1.5 in

Weight Approx. 195 g / < 1 lb (with batteries)

Delivery Scope

1 x Digital multimeter PCE-DM 5

2 x 1.5V AAA batteries

1 x Pair of test leads

1 x Thermocouple type-K

1 x Instruction manual

In addition to the measurement functions, the voltmeter can also generate a square-wave voltage of up to 10 kHz. Thanks to this function on the voltmeter, it can be checked, for example, whether the built-in frequency counters are working properly in industrial operation. The rubber cover on the voltmeter protects the measuring device from rough vibrations. Thanks to this rubber cover on the voltmeter, damage to the measuring equipment is prevented.

- TRMS measurements AC / DC

- NPN and PNP transistor testing

- Rubberized housing with tripod holder

- Generation of a quadratic function

- Automatic shutdown

- Continuity test and diode test

Specification

DC

Measuring range Resolution Accuracy

200 mV 100 µV ± (0.5% +5 digits)

2V 1 mV ± (0.5% +3 digits)

20V 10 mV ± (0.5% +3 digits)

200V 100 mV ± (0.5% +3 digits)

1000V 1V ± (1.0% +10 digits)

Input impedance: 10 MΩ

Surge protection: 1000V DC / AC RMS

AC

Measuring range Resolution Accuracy

2V 1 mV ± (0.8% +5 digits)

20V 10 mV ± (0.8% +5 digits)

200V 100 mV ± (0.8% +5 digits)

1000V 1V ± (1.2% +10 digits)

Input impedance: 10 MΩ

Surge protection: 1000V DC / AC RMS

The accuracies refer to 10 ... 100% of the respective measuring ranges, one

Frequency of 40 ... 400 Hz with a sine wave (True RMS) and a crest factor

(Crest factor): CF < 3. With a crest factor CF > 2 the increases

1% accuracy.

Capacity

Measuring range Resolution Accuracy

6 nF 1 pF ± (5.0% +40 digits)

60 nF 10 pF ± (2.5% +20 digits)

600 nF 100 pF ± (2.5% +20 digits)

6 µF 1 nF ± (2.5% +20 digits)

60 µF 10 nF ± (2.5% +20 digits)

600μF 100 nF ± (5.0% +10 digits)

6 mF 1 µF ± (5.0% +40 digits)

20 mF 10 µF ± (5.0% +40 digits)

Surge protection: 600V DC / AC RMS

The accuracies refer to 10 ... 100% of the respective measuring ranges

Response time for capacities> 1 mF approx. 8 seconds

The accuracy does not include the loading capacity

Transistor test

Measuring range Resolution Accuracy

0 ... 1000 hFE 1 hFE -

For NPN and PNP transistors

Base current approx. 10 µA

Collector-emitter voltage approx. 3V

Direct current

Measuring range Resolution Accuracy Voltage recording

20 µA 0.01 µA ± (1.2% +8 digits) 1.25 mV / µA

200 µA 0.1 µA ± (1.2% +8 digits) 0.125 mV / µA

2 mA 1 µA ± (1.2% +8 digits) 125 mV / mA

20 mA 10 µA ± (1.2% +8 digits) 3.75 mV / mA

200 mA 100 µA ± (1.2% +8 digits) 3.75 mV / mA

2 A 1 mA ± (1.5% +10 digits) 37.5 mV / A

20 A. 10 mA ± (2.0% +6 digits) 37.5 mV / A

Fuse up to the measuring range 200-mA: 200-mA / 250V

Fuses from a measuring range of 2 A: 20 A / 250V

Alternating current

Measuring range Resolution Accuracy Voltage recording

200-mA 100 µA ± (1.5% +15 digits) 3.75 mV / mA

2 A 1-mA ± (2.0% +5 digits) 37.5 mV / A

20 A. 10-mA ± (3.0% +10 digits) 37.5 mV / A

Fuse up to the measuring range 200-mA: 200-mA / 250V

Fuses from a measuring range of 2 A: 20 A / 250V

The accuracies refer to 10 ... 100% of the respective measuring ranges,

a frequency of 40 ... 400 Hz with a sine wave (True RMS) and one

Crest factor: CF < 3. With a crest factor CF > 2 the increases

1% accuracy.

Resistance

Measuring range Resolution Accuracy Short circuit current

200 Ω 0.1 Ω ± (0.8% +5 digits) approx.0.4-mA

2 kΩ 1 Ω ± (0.8% +3 digits) approx. 100 µA

20 kΩ 10 Ω ± (0.8% +3 digits) approx. 10 µA

200 kΩ 100 Ω ± (0.8% +3 digits) approx. 1 µA

2 MΩ 1 kΩ ± (0.8% +3 digits) approx. 0.2 µA

200 MΩ 100 kΩ ± (5% +30 digits) approx. 0.2 µA

Voltage with open circuit: approx. 1V

Surge protection: 600V DC / AC RMS

Diode test Test current Test voltage Accuracy

Measuring range Approx.0.4-mA Approx.3.3V 5%

Surge protection: 600 V DC / AC RMS

Continuity test Test current Test voltage Accuracy

Measuring range Approx.0.4-mA Approx.3.3V 5%

Surge protection: 600 V DC / AC RMS

Beep at 50 ± 20Ω

Function generation

Tension Function generation

(square) Approx.3.3V 50 Hz, 100 Hz, 200 Hz, 500

Hz, 1 kHz, 2 kHz

5 kHz, 10 kHz

Surge protection: 600V DC / AC RMS

Further specification

Display

LCD display

Largest display 1999 (3 1/2)

Measuring rate Approx. 3 measurements per second

Overrange "OL" display

Operating conditions 0 ... 40°C / 32 ... 104°F, < 75% RH non-condensing

Power supply 1 x 9V block battery

Dimensions

190 x 88.5 x 27.5 mm

Weight Approx. 320 g / < 1 lb (with battery)

Delivery Scope

1 x Digital multimeter PCE-DM 4

1 x Pair of test leads

1 x 9V block battery

1 x Instruction manual

The special feature of this digital multimeter is the L-conductor determination function. In this function, the L and N conductors can be distinguished from, for example, sockets with the digital multimeter. Only one measuring lead is needed for the digital multimeter. Once the L-wire has been broken, "LIVE" will be displayed on the Digital Multimeter. Another special feature of the digital multimeter is the cable finder function. With this function, live cables can be detected with the digital multimeter.

- Fast L-conductor determination

- Small and handy for quick use

- Many extensive measuring functions

- Backlit LC display

- Automatic shutdown

- Cable identification of live cables

Specifications

Voltage measurement DC

measuring range Resolution Accuracy

6V 1 mV ± (0.5% ± 3 Dgt.)

60V 10 mV ± (0.5% ± 3 Dgt.)

600V 100 mV ± (0.5% ± 3 Dgt.)

1000V 1V ± (0.5% ± 3 Dgt.)

Input resistance 10 MΩ

Maximum input voltage 1000V DC, 750V AC RMS

Voltage measurement AC

measuring range Resolution Accuracy

6V 1 mV ± (0.8% ± 3 Dgt.)

60V 10 mV ± (0.8% ± 3 Dgt.)

600V

100 mV ± (0.8% ± 10 Dgt.)

1000V 1V ± (0.8% ± 10 Dgt.)

Input resistance 10 MΩ

Maximum input voltage 1000V DC, 750V AC RMS

True RMS frequency range 45 ... 1000 Hz

Resistance

measuring range Resolution Accuracy

600 0.1 ± (0.8% ± 3 Dgt.)

6 k 1 ± (0.8% ± 3 Dgt.)

60 k 10 ± (0.8% ± 3 Dgt.)

600 k 100

± (0.8% ± 3 Dgt.)

6 m 1 k ± (0.8% ± 3 Dgt.)

60 m 10 k ± (1.2% ± 30 Dgt.)

Overvoltage protection

Measuring range

600V AC / DC

Measuring range Resolution

Diodes

0 ... 3.3V 0.001V Displays the diode forward

voltage

Overvoltage protection

Measuring range

600V AC / DC

Continuity test

Measuring range Resolution Accuracy

50 0,1Ω Beep at < 50 Ω

Measuring function

Capacity

Measuring range Resolution Accuracy

6 nF 0.001 nF ± (4% ± 10 Dgt.)

60 nF 0.01 nF ± (4% ± 5 Dgt.)

600 nF 0.1 nF ± (4% ± 5 Dgt.)

6 μF 1 nF ± (4% ± 5 Dgt.)

60 μF 10 nF ± (4% ± 5 Dgt.)

600 μF 100 nF ± (4% ± 5 Dgt.)

6 mF 1 μF ± (4% ± 5 Dgt.)

Overvoltage protection

Measuring range

600V AC / DC

Measuring function

Current measurement DC

Measuring range resolution accuracy

60-mA 0.01-mA ± (0.8% ± 3 Dgt.)

200-mA 0.1-mA ± (0.8% ± 3 Dgt.)

Maximum input current

Measuring range

200-mA AC / DC RMS

Current measurement in a frequency range between 40 ... 1 kHz True-Rms

Measuring function

Temperature measurement

Measuring range Resolution Accuracy

-20 ... 500°C 1°C ± (2% ± 2 Dgt.)

-4 ... 932°F 2°F ± (2% ± 2 Dgt.)

Overvoltage protection

Measuring range

600V AC / DC

Other specifications

Power supply battery 2 x 1.5V AA

Storage conditions -10 ... 50°C / 14 ... 122°F, < 75% rh

Dimensions 140 x 50 x 30 mm / 5.5 x 2 x 1.2 in

Weight About 119 g / < 1 lb

Delivery Scope

1 x Digital multimeter PCE-TT 4

1 x Pair of test lead

1 x Thermocouple type-K

1 x Transport bag

2 x 1.5V AAA batteries

1 x User manual

Another special feature of the multimeter is the mini USB interface. This allows the multimeter to be connected to any PC. LIVE data transmission from the multimeter can be carried out using the software supplied. Another difference between this and other multimeters is that this multimeter can be operated entirely via the keyboard instead of a measuring selector switch. As a result, this multimeter has fewer wear elements than conventional multimeters.

- Software for analyzing the measured values on the PC

- Numerical display with bar chart

- Current measurement up to 10 A for a short time

- Fully operable via the keyboard

- Type K thermocouples can be connected

- Voltage measurement up to 1000V.

Specification

DC

Measuring range Resolution Accuracy

600 mV

0.1 mV ± (0.5% + 4 digits)

6V 1 mV ± (0.5% + 4 digits)

60V 10 mV ± (0.5% + 4 digits)

600V 100 mV ± (0.5% + 4 digits)

1000V 1V ± (1.0% + 4 digits)

Input impedance for the measuring range 600 mV: > 60 MΩ

Input impedance for all other measuring ranges: 10 MΩ

Overload protection: 1000V DC / 750V AC peak value

AC

Measuring range Resolution Accuracy

6V 1 mV ± (0.8% + 10 digits)

60V 10 mV ± (0.8% + 10 digits)

600V 100 mV ± (0.8% + 10 digits)

750V 1V ± (1.0% + 6 digits)

Input impedance: 10 MΩ

Overload protection: 1000V DC / 750V AC peak value

Frequency range: 40 ... 400 Hz

Display duty cycle: 0.1 ... 99.9%

Alternating current

Measuring range Resolution Accuracy

60-mA 10 µA ± (1.5% + 10 digits)

600-mA 100 µA ± (1.5% + 10 digits)

6 A 1-mA ± (2.5% + 15 digits)

10 A 10-mA ± (2.5% + 15 digits)

Maximum input current: 10 A (for no longer than 10 seconds)

Fuses: 0.6 A / 250V, 10 A / 250V

Frequency range: 40 ... 400 Hz

Display duty cycle: 0.1 ... 99.9%

Resistance

Measuring range Resolution Accuracy

600 Ω 0.1 Ω ± (0.8% + 5 digits)

6 kΩ 1 Ω ± (0.8% + 4 digits)

60 kΩ 10 Ω ± (0.8% + 4 digits)

600 kΩ 100 Ω

± (0.8% + 4 digits)

6 MΩ 1 kΩ ± (0.8% + 4 digits)

60 MΩ 10 kΩ ± (1.2% + 10 digits)

Surge protection: 250V DC / 250V AC peak value

Measuring voltage open circuit: 600 mV

Capacity

Measuring range Resolution Accuracy

40 nF 10 pF ± (5.0% + 30 digits)

400 nF 100 pF ± (3.5% + 8 digits)

4 µF

1 nF ± (3.5% + 8 digits)

40 µF 10 nF ± (3.5% + 8 digits)

200 µF 100 nF ± (5.0% + 10 digits)

Surge protection: 250V DC / 250V AC peak value

Frequency

Measuring range Resolution Accuracy

10 Hz 0.01 Hz ± (0.3% + 2 digits)

100 Hz 0.1 Hz ± (0.3% + 2 digits)

1000 Hz 1 Hz ± (0.3% + 2 digits)

10 kHz 10 Hz ± (0.3% + 2 digits)

100 kHz 100 Hz ± (0.3% + 2 digits)

1 MHz 1 kHz ± (0.3% + 2 digits)

20 MHz 10 kHz ± (0.3% + 2 digits)

Input sensitivity: 1.5V

Surge protection: 250V DC / 250V AC peak value

Temperature measurement thermocouple type K

Measuring range Resolution Accuracy

-20 ... 400°C 0.1°C ± (1.0% + 50 digits)

400 ... 1000°C 1°C ± (1.5% + 15 digits)

0 ... 750°C 0.1°F ± (1.0% + 50 digits)

750 ... 1832° F 1°F ± (1.5% + 15 digits)

Diode test Test current Test voltage Accuracy

1-mA 3V -

Surge protection: 250V DC / 250V AC peak value

Continuity test Test current Test voltage Accuracy

- 1.2V -

Surge protection: 250V DC / 250V AC peak value

Beep at < 30 Ω

More specifications

Display LCD display

Largest display 5999 3 5/6

Bar graph Scaled in 61 bars

Measuring rate 3 measurements per second

Overrange "OL" is displayed

Data Interface Mini USB

Battery status indicator With a total battery voltage of 2.4V

Operating conditions 0 ... 40°C / 32 ... 104°F

Storage conditions -10 ... 50°C / 14 ... 122°F, < 80% RH

Power supply 2 x 1.5V AAA batteries

Dimensions 185 x 91 x 49 mm / 7.3 x 3.6 x 1.9 in

Weight Approx. 410 / < 1 lb (with batteries)

Delivery Scope

1 x Digital Multimeter PCE-DM 15

1 x Pair of test leads

1 x Type K thermocouple

2 x 1.5V AAA batteries

1 x Mini USB cable

1 x Software CD

1 x Instruction manual

Simple and direct selection of the basic functions by means of pushbuttons and autom. Range selection for: DCV, ACV, DCA, ACA, ohms, diode, continuity, capacitance, frequency and temperature measurement functions. The digital multimeter measures voltages up to 1000V AC / DC. The current can be measured directly up to 10 A AC / DC and over the Rogowski coil up to 3000 A AC. The case of the digital multimeter is IP67 protected and thus perfectly usable in a robust everyday industrial environment.

- OLED display

- Minimum value and maximum value, peak level

- Hold function

- Relative value measurement

- Diode test

- Continuity tester

- Incl. current loop

- Automatic range selection

- Automatic shutdown

- Incl. Rogowski coil for 3000 A AC

Specifications

Measuring function

DC voltage DC

Measuring range Resolution Accuracy

60 mV 0.01 mV ± (0.5% from + 5 Dgt)

600 mV 0.1 mV ± (0.5% from + 5 Dgt)

6V 0.001 V ± (0.5% from + 5 Dgt)

60V 0.01V ± (0.5% from + 5 Dgt)

600V 0.1V ± (0.8% from + 5 Dgt)

1000V 1 V ± (0.8% from + 5 Dgt)

AC voltage AC Measuring range Resolution Accuracy

(TRMS) 60 mV 0.01 mV @ 50 ... 60 Hz:

600 mV 0.1 mV ± (1% from + 5 Dgt)

6V 0.001V

60V 0.01V @ 61Hz ... 1KHz:

600V 0.1V ± (3% from + 5 Dgt)

1000V 1V

AC + DC TRMS Measuring range Resolution Accuracy

Tension 6V 0.001V @ 50Hz ... 1KHz:

60V 0.01V ± (3% from + 40 Dgt)

600V 0.1V

1000V 1V

DC DC Measuring range Resolution Accuracy

600 μA 0.1 μA ± (1% from + 5 Dgt)

6000 μA 1 μA ± (1% from + 5 Dgt)

60-mA 0.01-mA ± (1% from + 5 Dgt)

600-mA 0.1-mA ± (1% from + 8 Dgt)

6 A 0.001 A ± (1.5% from + 8 Dgt)

10 A 0.01A ± (1.5% from + 8 Dgt)

AC AC Measuring range Resolution Accuracy

(TRMS) 600 μA 0.1 μA ± (1.2% from + 5 Dgt)

6000 μA 1 μA ± (1.2% from + 5 Dgt)

60-mA 0.01-mA ± (1.2% from + 5 Dgt)

600-mA 0.1-mA ± (1.2% from + 5 Dgt)

6 A 0.001 A ± (1.5% from + 8 Dgt)

10 A 0.01A ± (1.5% from + 8 Dgt)

4 ... 20-mA % -25 ... 125% 0.01% ± 50 digits

AC AC measuring range resolution accuracy

Over Rogowski coil 30 A 0.01A ± (1.2% vMw + 10 Dgt)

300 A 0.1 A

3000 A 10 A

Resistance Measuring range Resolution Accuracy

600 Ω 0.1 Ω ± (1.2% vMw + 10 Dgt)

6 kΩ 0.001 kΩ ± (1.2% vMw + 5 Dgt)

60 kΩ 0.01 kΩ ± (1.2% vMw + 5 Dgt)

600 kΩ 0.1 Ω ± (1.2% vMw + 5 Dgt)

6 MΩ 0.001 MΩ ± (1.2% vMw + 5 Dgt)

60 MΩ 0.01 MΩ ± (2.5% vMw + 10 Dgt)

Frequency Measuring range Resolution Accuracy

40 Hz ... 100 kHz 0.01 Hz ... 0.001 kHz ± 0.5% of Mw.

Duty cycle Measuring range Resolution Accuracy

20 ... 80% 0.1% ± (1.2% from + 2 Dgt)

Capacity Measuring range Resolution Accuracy

999.9 nF 0.1 nF ± (1.2% from + 8 Dgt)

9.999 nF 0.001 μF ± (1.5% from + 8 Dgt)

999.9 nF 0.1 μF ± (1.5% from + 8 Dgt)

99.99 n.f. 0.01 mF ± (2.5% vMw + 20 Dgt)

Temperature with Measuring range Resolution Accuracy

Thermocouple type K -40 ... 600°C / -40 ... 1112°F 0.1°C / 0.18°F ± (1.5% from + 3°C / 5.4°F)

600 ... 1000°C / 1112 ... 1832°F 1°C / 1.8°F ± (1.5% from + 3°C / 5.4°F)

General technical data

Isolation Class 2 Double insulation

Casing IP67 plastic housing

Diode test 1.5-mA test current

3V DC open circuit voltage

Continuity test Alarm tone if resistance < 50 Ω

Display 6,000 digit OLED

Refresh rate display 3 Hz

Automatic shutdown After 30 minutes

Input impedance 30 MΩ AC / DC V

AC response TRMS

AC bandwidth 50 ... 1000 Hz

Power supply Digital multimeter: 9V block battery

Rogowski coil: 2 x 1.5V AAA battery

Fuses 800-mA / 1000V fast acting (6.3 x 32 mm)

10 A / 1000V fast acting (10 x 38 mm)

Operating conditions 0 ... 40°C / 32 ... 104°F, < 70% rh

Storage conditions -10 ... 60°C / 14 ... 140°F, < 80% rh

Operating altitude < 2000 m / 6561.7 ft above sea level.

Dimensions 170 x 79 x 50 mm / 6.7 x 3.1 x 2 in

Weight 342 g / < 1 lb

Security UL61010-1 v3

CAT IV 600V

CAT III 1000V

Pollution degree

2

Delivery Scope

1 x Digital Multimeter PCE-ODM 10

1 x Rogowski coil PCE-DM-CLAMP

1 x Set of test leads

1 x Thermocouple type K

1 x Thermocouple adapter

1 x Carrying bag

1 x 9V block batteries

2 x 1.5V AAA batteries

1 x User manual

Simple and direct selection of the basic functions by means of pushbuttons and autom. Range selection for: DCV, ACV, DCA, ACA, ohms, diode, continuity, capacitance, frequency and temperature measurement functions. The digital multimeter measures voltages up to 1000V AC / DC. The current can be measured directly up to 10 A AC / DC and over the Rogowski coil up to 3000 A AC. The case of the digital multimeter is IP67 protected and thus perfectly usable in a robust everyday industrial environment.

- OLED display

- Minimum value and maximum value, peak level

- Hold function

- Relative value measurement

- Diode test

- Continuity tester

- Incl. current loop

- Automatic range selection

- Automatic shutdown

- Incl. Rogowski coil for 3000 A AC

- Bluetooth

Specification

Measuring function

DC voltage DC

Measuring range Resolution Accuracy

60 mV 0.01 mV ± (0.5% from + 5 Dgt)

600 mV 0.1 mV ± (0.5% from + 5 Dgt)

6V 0.001 V ± (0.5% from + 5 Dgt)

60V 0.01V ± (0.5% from + 5 Dgt)

600V 0.1V ± (0.8% from + 5 Dgt)

1000V 1 V ± (0.8% from + 5 Dgt)

AC voltage AC Measuring range Resolution Accuracy

(TRMS) 60 mV 0.01 mV @ 50 ... 60 Hz:

600 mV 0.1 mV ± (1% from + 5 Dgt)

6V 0.001V

60V 0.01V @ 61Hz ... 1KHz:

600V 0.1V ± (3% from + 5 Dgt)

1000V 1V

AC + DC TRMS Measuring range Resolution Accuracy

Tension 6V 0.001V @ 50Hz ... 1KHz:

60V 0.01V ± (3% from + 40 Dgt)

600V 0.1V

1000V 1V

DC DC Measuring range Resolution Accuracy

600 μA 0.1 μA ± (1% from + 5 Dgt)

6000 μA 1 μA ± (1% from + 5 Dgt)

60-mA 0.01-mA ± (1% from + 5 Dgt)

600-mA 0.1-mA ± (1% from + 8 Dgt)

6 A 0.001 A ± (1.5% from + 8 Dgt)

10 A 0.01A ± (1.5% from + 8 Dgt)

AC AC Measuring range Resolution Accuracy

(TRMS) 600 μA 0.1 μA ± (1.2% from + 5 Dgt)

6000 μA 1 μA ± (1.2% from + 5 Dgt)

60-mA 0.01-mA ± (1.2% from + 5 Dgt)

600-mA 0.1-mA ± (1.2% from + 5 Dgt)

6 A 0.001 A ± (1.5% from + 8 Dgt)

10 A 0.01A ± (1.5% from + 8 Dgt)

4 ... 20-mA % -25 ... 125% 0.01% ± 50 digits

AC AC measuring range resolution accuracy

Over Rogowski coil 30 A 0.01A ± (1.2% vMw + 10 Dgt)

300 A 0.1 A

3000 A 10 A

Resistance Measuring range Resolution Accuracy

600 Ω 0.1 Ω ± (1.2% vMw + 10 Dgt)

6 kΩ 0.001 kΩ ± (1.2% vMw + 5 Dgt)

60 kΩ 0.01 kΩ ± (1.2% vMw + 5 Dgt)

600 kΩ 0.1 Ω ± (1.2% vMw + 5 Dgt)

6 MΩ 0.001 MΩ ± (1.2% vMw + 5 Dgt)

60 MΩ 0.01 MΩ ± (2.5% vMw + 10 Dgt)

Frequency Measuring range Resolution Accuracy

40 Hz ... 100 kHz 0.01 Hz ... 0.001 kHz ± 0.5% of Mw.

Duty cycle Measuring range Resolution Accuracy

20 ... 80% 0.1% ± (1.2% from + 2 Dgt)

Capacity Measuring range Resolution Accuracy

999.9 nF 0.1 nF ± (1.2% from + 8 Dgt)

9.999 nF 0.001 μF ± (1.5% from + 8 Dgt)

999.9 nF 0.1 μF ± (1.5% from + 8 Dgt)

99.99 n.f. 0.01 mF ± (2.5% vMw + 20 Dgt)

Temperature with Measuring range Resolution Accuracy

Thermocouple type K -40 ... 600°C / -40 ... 1112°F 0.1°C / 0.18°F ± (1.5% from + 3°C / 5.4°F)

600 ... 1000°C / 1112 ... 1832°F 1°C / 1.8°F ± (1.5% from + 3°C / 5.4°F)

General technical data

Isolation Class 2 Double insulation

Casing IP67 plastic housing

Diode test 1.5-mA test current

3V DC open circuit voltage

Continuity test Alarm tone if resistance < 50 Ω

Display 6,000 digit OLED

Refresh rate display 3 Hz

Automatic shutdown After 30 minutes

Input impedance 30 MΩ AC / DC V

AC response TRMS

AC bandwidth 50 ... 1000 Hz

Interface Bluetooth BLE

Power supply Digital multimeter: 9 V block battery

Rogowski coil: 2 x 1.5V AAA battery

Fuses 800-mA / 1000V fast acting (6.3 x 32 mm)

10 A / 1000V fast acting (10 x 38 mm)

Operating conditions 0 ... 40°C / 32 ... 104°F, < 70% rh

Storage conditions -10 ... 60°C / 14 ... 140°F, < 80% rh

Operating altitude < 2000 m / 6561.7 ft above sea level.

Dimensions 170 x 79 x 50 mm / 6.7 x 3.1 x 2 in

Weight 342 g / < 1 lb

Security UL61010-1 v3

CAT IV 600V

CAT III 1000V

Pollution degree

2

Delivery Scope

1 x Digital Multimeter PCE-ODM 12

1 x Rogowski coil PCE-DM-CLAMP

1 x Set of test leads

1 x Thermocouple type K

1 x Thermocouple adapter

1 x Carrying bag

1 x 9V block batteries

2 x 1.5V AAA batteries

1 x User manual

Thanks to the cable finder function of the voltage tester, the operator can find the required cable in no time at all. To do this, the voltage tester must be connected to the connection of a cable. The voltage tester now acts as a transmitter. With the supplied receiver you can now easily find the desired cable in, for example, a wiring harness or behind walls with the voltage tester.

To check whether a voltage is applied to a cable, the voltage tester has an NCV (Not Connected Voltage) function. With this function, the voltage tester must only be held to the line. Both acoustically and visually, the voltage tester signals the operator that a voltage is applied to the line. Thanks to this contactless measuring function, the voltage tester reduces accidents at work due to electric shock.

- Extensive network connection testing

- Telephone connection examination

- Acoustic cable finder function

- Robust housing for mobile use

- Voltage, current, resistance measurement

- Contactless voltage detection

Specification

Measuring function

AC

Measuring range Resolution Accuracy

200 mV 0.1 mV ± (0.8% + 5 digits)

2V 1 mV ± (0.8% + 5 digits)

20V 0.01V ± (0.8% + 5 digits)

200V 0.1V ± (0.8% + 5 digits)

750V 1V

± (0.8% + 5 digits)

Maximum input voltage: 750V AC

Frequency range: 40 ... 400 Hz

Conversion RMS

DC

Measuring range Resolution Accuracy

200 mV 0.1 mV ± (0.5% + 5 digits)

2V 1 mV ± (0.5% + 5 digits)

20V 0.01V ± (0.5% + 5 digits)

200V 0.1V ± (0.5% + 5 digits)

1000V 1V ± (0.8% + 5 digits)

Maximum input voltage: 1000V DC

Alternating current

Measuring range Resolution Accuracy

200 μA 0.1 μA

± (1.5% + 5 digits)

2000 μA 1 μA ± (1.5% + 5 digits)

20-mA 0.01-mA ± (1.5% + 5 digits)

200-mA 0.1-mA ± (1.5% + 5 digits)

2 A 0.001 A ± (3.0% + 5 digits)

20 A 0.01A ± (3.0% + 5 digits)

Overload protection at μA, mA: 200-mA / 1000V fuse

Overload protection at A: 10 A / 500V fuse

maximum input current μA: 2000 μA

maximum input current mA: 200-mA

maximum input current A: 10 A.

Frequency range: 40 ... 400 Hz

Conversion RMS

Measuring function

Direct current

Measuring range Resolution Accuracy

200 μA 0.1 μA ± (1.2% + 5 digits)

2000 μA 1 μA ± (1.2% + 5 digits)

20-mA 0.01-mA ± (1.2% + 5 digits)

200-mA 0.1-mA ± (1.2% + 5 digits)

2 A 0.001 A ± (2.0% + 5 digits)

20 A 0.01A ± (2.0% + 5 digits)

Overload protection at μA, mA: 200-mA / 1000V fuse

Overload protection at A: 10 A / 500V fuse

maximum input current μA: 2000 μA

maximum input current mA: 200-mA

maximum input current A: 10 A.

Measuring function

Resistance

Measuring range Resolution Accuracy

200 Ω 0.1 Ω ± (1.0% + 5 digits)

2 kΩ 0.001 kΩ ± (1.0% + 5 digits)

20 kΩ 0.01 kΩ ± (1.0% + 5 digits)

200 kΩ 0.1 kΩ ± (1.0% + 5 digits)

2 MΩ 0.001 MΩ ± (1.0% + 5 digits)

20 MΩ 0.01 MΩ ± (1.0% + 5 digits)

Diode measurement

Measuring range Resolution Accuracy

Indicates the forward voltage 1 mV -

Continuity test

Measuring range Resolution Accuracy

Beep at <70 Ω 0.1 Ω -

Other specifications

Range selection Automatically and manually

Measurement with receiver Detecting cable bridges, detecting cable runs

Measurement with network function Detection of open and closed circuits,

wrong connection, "split pair", "reverse" connection,

Network integration test, screening test

Further measurement function "NCV" detection, telephone connection testing

Display LCD, 1999 digits

Overload indicator "OIL"

Automatic shutdown After 15 minutes of inactivity

Fuses F10A / 500V, F200mA / 1000A

Battery indicator Display when battery is discharged

Power supply 9V block battery

Operating conditions 0 ... 40°C / 32 ... 104°F

-10 ... 50°C / 14 ... 122°F

Dimensions 195 x 92 x 55 mm / 7.7 x 3.6 x 2.2 in

Weight Approx. 400 g / < 1 lb (with batteries)

Delivery Scope

1 x Digital multimeter PCE-LT 12

1 x Receiver

1 x Pair of test leads

1 x Connection adapter strand

2 x 9V block batteries

1 x User manual

The multimeter is equipped with a Bluetooth interface, which can be read out via an app. In the app, various visualizations and a data logger are available. In order to be able to work with the digital multimeter even in low visibility and light conditions, the multimeter is equipped with a measuring point illumination and a display illumination.

- Capacity measurement

- MAX function

- Measuring point and display illumination

- Temperature measurement via thermocouple type K

- Data logger function via app

- Diode test

Specifications

Display 4000 digit LCD

Isolation double, class II

Casing Double molded body, IP67 (waterproof and dustproof)

Diode test Test current maximum 1.5-mA

no-load voltage 3V

Continuity test Limit <50 Ω

Overload indicator Display of "OL" in the display

Battery indicator Display at low battery voltage

Polarity Display of "-" in negative polarity

Measuring rate 2 measurements per second, nominal

Automatic shutdown after about 15 minutes of inactivity

Input resistance 10 MΩ AC / DC

AC coupling TRMS (50 ... 400 Hz)

Bandwidth 50 ... 400 Hz

Power supply 3 x 1.5V AAA battery

Fuse 10 A / 600V (5 x 20 mm)

Operating conditions 0 ... 40°C / 32 ... 104°F, <70% rel. humidity

Storage conditions -10 ... 60°C / 14 ... 140°F, <80% rel. humidity

Operating altitude 2000 m / 6562 ft

Dimensions 147 x 68 x 50 mm / 5.8 x 2.7 x 2 in

Weight 318 g / < 1 lb

Security UL 61010-1 V-3

CAT III 600V Degree of

pollution 2

Measuring range Resolution Accuracy (% v.Mw.)

Alternating current 4 A 1-mA ± 2.5% + 5 Dgt

10 A 10-mA ± 2.5% + 5 Dgt

Overload protection: 10 A / 600V fuse

AC coupling: 50 ... 60 Hz

Direct current 4 A 1-mA ± 2% + 3 Dgt

10 A 10-mA ± 2% + 3 Dgt

Overload protection: 10 A / 600V fuse

AC 4V 1 mV ± 1% + 5 Dgt

40V 10 mV ± 1.2% + 5 Dgt

400V 0.1V ± 1.5% + 5 Dgt

600V 1V ± 1.5% + 5 Dg

Input protection: 600V DC or 600V AV RMS

Input resistance: 10 MΩ

AC Coupling: 50 ... 60 Hz

AC 4V 1 mV ± 1.5% + 5 Dgt

(low input resistance) 40V 10 mV ± 1.5% + 5 Dgt

400V 0.1V ± 1.5% + 5 Dgt

All measuring ranges of the alternating voltage were specified with 5 ... 100% of the measuring range

Input protection: 300V AC RMS or 300V DC

Input resistance: approx. 3 kΩ

AC coupling: 50 ... 60 Hz

DC 400 mV 0.1 mV ± 1% + 8 Dgt

4 V 1 mV ± 1% + 3 Dgt

40V 10 mV ± 1% + 3 Dgt

400V 0.1V ± 1% + 3 Dgt

600V 1V ± 1.2% + 3 Dgt

Input protection: 600V DC or 600V AV RMS

Input resistance: 10 MΩ

DC 400 mV 0.1 mV ± 1.5% + 5 Dgt

(low input resistance) 4 V 1 mV ± 1.5% + 5 Dgt

40V 10 mV ± 1.5% + 5 Dgt

400V 0.1V ± 1.5% + 5 Dgt

Input protection: 300V AC RMS or 300V DC

Input resistance: approx. 3 kΩ

Resistance 400 Ω 0.1 Ω ± 1.5% + 5 Dgt

4 kΩ 1 Ω ± 1.5% + 5 Dgt

40 kΩ 10 Ω ± 1.5% + 5 Dgt

400 kΩ 100 Ω ± 1.5% + 5 Dgt

4 MΩ 1 kΩ ± 2% + 10 Dgt

40 MΩ 10 kΩ ± 2% + 10 Dgt

Input protection: 600V AC RMS or 600V DC

Capacity 40 nF 0.01 pF ± 5% + 35 Dgt

400 nF 0.1 pF ± 3% + 5 Dgt

4 μF 0.001 μF ± 3% + 5 Dgt

40 μF 0.01 μF ± 3% + 5 Dgt

400 μF 0.1 μF ± 3% + 5 Dgt

4000 μF 1 μF ± 5% + 5 Dgt

Input protection: 600V AC RMS or 600V DC

Frequency 9,999 Hz 0.001 Hz ± 1% + 5 Dgt

99.99 Hz 0.01 Hz ± 1% + 5 Dgt

999.9 Hz 0.1 Hz ± 1% + 5 Dgt

9,999 kHz 1 Hz ± 1% + 5 Dgt

99.99 kHz 10 Hz ± 1% + 5 Dgt

999.9 kHz 100 Hz ± 1.2% + 5 Dgt

9,999 MHz 1 kHz ± 1.2% + 5 Dgt

Input protection: 600V AC RMS or 600V DC

Sensitivity:> 8V RMS

Duty cycle 1 ... 99% 0.1% ± 1.2% + 2 Dgt

Input protection: 600V AC RMS or 600V DC

Pulse width: 0.1 ... 100 mS

Frequency range: 5 Hz ... 10 KHz

Sensitivity: > 8V RMS

Temperature (type K thermocouple) -18 ... 760°C / -0.4 ... 1400°F 0.1°C / 0.18°F ± 2% + 5°C / 9°F

Input protection: 300V AC RMS or 300V DC

Delivery Scope

1 x Digital Multimeter PCE-HDM 7

2 x Measuring line

2 x 1.5V battery

1 x Storage bag

1 x Thermocouple type K

1 x Thermocouple adapter

1 x User manual